Foxboro d/p Cell Transmitters have been the standard of process (a) See Optional Capsule Materials (13A Series Only) for span limits of 13A4 (Monel capsule. The Foxboro® brand Models 15A, 13A, and 13HA pneumatic d/p Cell transmitters measure differential pressure and. Download Manual Model. Regarding This Manual. • This manual should be provided to the end user. • The contents of this manual are subject to change without prior.
|Country:||Antigua & Barbuda|
|Published (Last):||3 February 2012|
|PDF File Size:||20.99 Mb|
|ePub File Size:||17.79 Mb|
|Price:||Free* [*Free Regsitration Required]|
Models 15A, 13A, and 13HA Pneumatic d/p Cell® Transmitters
All the reading obtained will be the as found readings. In most calibration done, you will be doing either a bench shop calibration — A bench calibration is a procedure where the device is calibrated at a calibration bench using calibration devices to simulate the process, — or a foxbpro calibration where the actual process is used.
The graph below illustrates the correlation between input and output values. A typical DP cell transmitter calibration diagram is shown below:.
Models 15A, 13A, and 13HA Pneumatic d/p Cell® Transmitters | Schneider Electric
The potentiometer has a maximum of 20 turns between minimum and maximum resistance therefore turning the ZERO or SPAN screws clockwise or counter-clockwise for 20 turns will cause the potentiometer to be at either maximum or minimum. Most transmitter calibration done is a five point calibration.
If you calibrate the DP transmitter before first testing and recording the as found data, the history of the device performance data will be lost. Connect all the equipment needed for coxboro calibration exercise in the appropriate manner. How to Calibrate Smart Transmitters. Readings are taken for both increasing and decreasing input values and the corresponding transmitter output values are recorded Step 5: You are now ready to calibrate the DP transmitter.
Foxboro 13A, 13HA, 15A Pneumatic D/P Cell Transmitter
Then connect them according to your connection diagram. Record the current reading in m A which will be your first data point. For your application, this could be modified slightly.
Note that these screws each connect to a variable resistance potentiometer and can be turned indefinitely. That is the potentiometer is of a type that once fully adjusted clockwise or counter clockwise the screw may continue to turn without further varying the resistance for either direction.
So once your equipment is well setup, power it up and pressurize the high port 13s your DP transmitter. Newer Post Older Post Home.
CalibrationDP Transmitters. However, if you are doing field calibration that requires the use of the actual process pressure, you will need a pressure regulator in conjunction with a pneumatic calibrator to help you control the pressure applied to the DP cell. For example if the pressure source is a hand pump, you can easily control the pressure applied to the DP cell.
Whether you are doing a bench calibration or a field calibration, the low port of the DP transmitter cell is vented to the atmosphere as shown in the connection diagram above and the high port of the DP transmitter connected to a pressure source e.
To calibrate an instrument involves checking that the output of the given instrument corresponds to given inputs at several points throughout the calibration range of the instrument.
That is, adjusting one has an effect on the other. Share to Twitter Share to Facebook. Record and put down the following can easily be sourced from transmitter nameplate: