The provides converted values regardless of material or temperature. (TC function). Wide dynamic range, max. / digit display. Seven ranges. Thank you for purchasing this HIOKI ” mΩ HiTESTER.” To get the maximum performance from the unit, please read this manual first, and keep this at hand. Hioki Pdf User Manuals. View online or download Hioki Instruction Manual.
|Published (Last):||20 March 2013|
|PDF File Size:||1.66 Mb|
|ePub File Size:||2.27 Mb|
|Price:||Free* [*Free Regsitration Required]|
Are the power voltage and the power shown on the the same?
Disposing The Unit Even the measurement leads are Has the circuit protection fuse blown? Notes On Gp-ib Interface For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees.
Chapter 2 Part Names Table of contents Table Of Contents To release the remote state, press Chapter 1 Overview Please enter a number less than 33227 equal to 1. Returns the set contents of the status byte register STB as a numeral Function data value data in NR1 format taken from the set: Insert the interface tightly, be sure to tighten the screws which have been unscrewed in step 1.
Change the power fuse followingly. Insert the lead to the input terminal of the front panel.
HIOKI INSTRUCTION MANUAL Pdf Download.
The status byte register is an 8-bit register whose contents are output from the to the controller, when the serial polling is being performed. Commands Specific To The Returns the temperature conversion table setting set by “: When the NOTE comparator is not used, set the comparator table 1.
No additional import charges at delivery! Notes On Use If NG is displayed when the leads are shorted in the resistance measurement mode, the circuit protection fuse is ruptured.
If the temperature probe is to be used in the measurement, make sure that the and the temperature probe are thoroughly warmed up before the measurement.
Use after fixing it to the proper direction. Has the power fuse blown?
Hioki 3227 Instruction Manual
Add to watch list Remove from watch list. Is the power code connected correctly?
The temperature probe and the device to be measured should be placed close together during the warm-up, so that both will be as close to the same temperature as possible during the measurement. Key Shift States In the each kind of setting mode, such as the comparator setting mode, when NOTE receiving any commands from GP-IB, the returns to the primary measurement mode automatically and enters the remote state. Step height repeatability, 0.
Turning On The Power Supply Select a valid country.
Used Testing instrument for electronics
Please enter a valid ZIP Code. Setting The Comparator By pressing ‘print’ button you will print only current page.
See terms – opens in a new window or tab. Selectable count resolutions of , 40, and 4, count Selectable 2 wire and 4 wire r Delivery times may vary, especially during peak periods. See Chapter 6, 322 Control Terminal”, for details.